Bioactive versus bare platinum coils in the treatment of intracranial aneurysms: the MAPS (Matrix and Platinum Science) trial
Department of Radiology
Nervous System Diseases | Neurology | Radiology
BACKGROUND AND PURPOSE: The ability of polymer-modified coils to promote stable aneurysm occlusion after endovascular treatment is not well-documented. Angiographic aneurysm recurrence is widely used as a surrogate for treatment failure, but studies documenting the correlation of angiographic recurrence with clinical failure are limited. This trial compares the effectiveness of Matrix(2) polyglycolic/polylactic acid biopolymer-modified coils with bare metal coils and correlates the angiographic findings with clinical failure (ie, target aneurysm recurrence), a composite end point that includes any incident of posttreatment aneurysm rupture, retreatment, or unexplained death.
MATERIALS AND METHODS: This was a multicenter randomized noninferiority trial with blinded end point adjudication. We enrolled 626 patients, divided between Matrix(2) and bare metal coil groups. The primary outcome was target aneurysm recurrence at 12 +/- 3 months.
RESULTS: At 455 days, at least 1 target aneurysm recurrence event had occurred in 14.6% of patients treated with bare metal coils and 13.3% of Matrix(2) (P = .76, log-rank test) patients; 92.8% of target aneurysm recurrence events were re-interventions for aneurysms that had not bled after treatment, and 5.8% of target aneurysm recurrence events resulted from hemorrhage or rehemorrhage, with or without retreatment. Symptomatic re-intervention occurred in only 4 (0.6%) patients. At 455 days, 95.8% of patients with unruptured aneurysms and 90.4% of those with ruptured aneurysms were independent (mRS ≤ 2). Target aneurysm recurrence was associated with incomplete initial angiographic aneurysm obliteration, presentation with rupture, and a larger aneurysmal dome and neck size.
CONCLUSIONS: Tested Matrix(2) coils were not inferior to bare metal coils. Endovascular coiling of intracranial aneurysms was safe, and the rate of technical success was high. Target aneurysm recurrence is a promising clinical outcome measure that correlates well with established angiographic measurements.
DOI of Published Version
AJNR Am J Neuroradiol. 2014 May;35(5):935-42. Epub 2014 Jan 30. Link to article on publisher's site
AJNR. American journal of neuroradiology
McDougall, C. G.; Johnston, S. Claiborne; Gholkar, A.; Barnwell, S. L.; Vazquez Suarez, J. C.; Masso Romero, J.; Chaloupka, J. C.; Bonafe, A.; Wakhloo, Ajay K.; Tampieri, D.; Dowd, C. F.; Fox, A. J.; Imm, S. J.; Carroll, K.; and Turk, A. S., "Bioactive versus bare platinum coils in the treatment of intracranial aneurysms: the MAPS (Matrix and Platinum Science) trial" (2014). Radiology Publications and Presentations. 268.